When should solutions be changed?
Cleaning solutions should be replenished when a noticeable decrease in cleaning action occurs, or when the solution is visibly dirty or spent. A fresh batch of solution at each cleaning session is usually not required.
Why must I keep solution at the tank's level indicator?
The solution level should always be maintained at the level indicator in the tank, with trays or beakers installed. The ultrasonic cleaning system is a 'tuned' system. Improper solution levels will change the characteristics of the environment, can affect the system frequency, decrease effectiveness, and potentially damage the cleaner. Maintaining the proper solution level provides optimum circulation of solution around parts, and protects heaters and transducers from overheating or stress.
What is the length of cleaning time?
Cleaning time will vary, depending on such things as soil, solution, temperature and the degree of cleanliness desired. Highly visible removal of soils should start almost immediately after the ultrasonic cleaning action begins. Cleaning time adjustment is the easiest (and most often misapplied) factor used to compensate for process variables. Although new application cycle an experienced operator can approximate duration, it usually must be validated by actual use with the chosen solution and the actual soiled parts.
How do I perform the "foil" test?
Cut three small pieces of aluminum foil about 4" x 8" each. Fold each piece over a rod that you will use to suspend the foil in the tank. A clothes hanger works well. Your cleaner should be filled with an ultrasonic cleaning solution, degassed, and brought up to normal operating temperature. Suspend the first "square" in the center of the tank and the other two a couple of inches from each end of the tank. Make sure that the tank is filled to the fill line, and turn on the ultrasonics for about ten minutes. Remove the foil and inspect: All three pieces of aluminum foil should be perforated and wrinkled to about the same degree.
Why must trays or beakers be used?
Items being cleaned should never be placed directly on the tank bottom. Transducers (which produce the ultrasound) are bonded to the bottom of the tank. Items resting directly on the tank bottom can damage the transducers and/or reduce cavitation. Additionally, a tray or beaker will position the item within the optimal cleaning zone of the tank. The tray or beaker will also hold the load together and allow for easy, no-touch removal, draining and transport of the items to the next step in the cleaning process.
What is the optimum cleaning temperature?
Heat usually enhances and speeds up the cleaning process, and most detergent solutions are designed to work best at an elevated temperature. The best way to find the optimum temperature, which will give you the fastest, cleanest and safest results, is to run tests. Usually, the best results are within the 50¡C to 65¡C range.
Is rinsing required after cleaning cycles?
Rinsing is recommended to remove any chemical residue, which could be harmful to the part. Parts can be rinsed right in your ultrasonic cleaner, using a clean water bath, or in a separate tub containing tap, distilled or de-ionized water
For more Information:Contact Us